I had 3 of these in the past years. I gave one away.
But the first one I used just a short time, and then the DC-voltage showed about 1,4 times (sqr(2)?) the real value. I easily bought a new one, because they have allways been rediculously cheap.
Recently the second one also got the same voltage error disease...
Does this sound familiar to some of you?
I opened both of them, but there is not much inside that looks reparable to me.
I like these multimeters because they are compact. May be someone has a repair solution?
< Merged the 2 topics - Plons >